The technology is based on a modified triboelectric principle detecting particles interacting with the
sensing rod and such particles just passing the rod. Build up on the rod surface will not be detected,
only moving particles generate a flow rate proportional signal which is monitored by the electronic.
Installation is done on the clean gas side downstream the filter at a metal duct by welding on of
a thread bush boring through the duct wall and screwing in dust watch. On and off distance should
this 3-fold of the pipe diameter area, the sensor length 1/3 to 2/3 of the pipe diameter.
The dust monitor FlowSwitch 700E is used for the detection of filter failure functions e.g. crack or defect in assembling.
There are no reviews yet.